Click on image to enlarge
Have you seen this happen before?
This is my 3rd time.
Plz respond in the comments section below- or send an email privately to firstname.lastname@example.org to share your experience and identify if this is indeed a common issue.
We are developing a database of common or not-so-common problems encountered by perfusionists during CPB.
Please join us by clicking this Link if you have had any sort of unusual event you would like to share. Please email details if you feel that your story will help other perfusionists troubleshoot similar issues while on bypass…
To View all submitted cases relating to Perfusion Problem Solving: Click Here
or click the “Intnl. CQI Registry” Link on the Top menu Bar.
- Water line leaked after attaching to Oxygenator. This is the 3rd time I have seen this. Observed what appears to be a mold deformity on the H2O outlet from the Oxygenator, distorted inwardly so that a water tight fit is not possible.
How Was The Problem Identified?
- Watched the connection leak upon connection- and spray out once the Heater/Cooler was turned on..
What Steps Were Taken ?
- Oxynenator was replaced.
- New Oxygenator looked normal and performed the Water Test well. Issue Resolved
What Clues Were Missed?
- Third time I have seen this happen.
- No issues.
- Manufacturer contacted
Want to Share a Problem or Unique Perfusion Issue?
(Click Image to View Summary of Glitches)
Catch the latest Perfusion news or peruse our article archive.